These equipments are used for climatic & durability tests of electrical & electronic components, corrosion test on mechanical assemblies, materials for simulated tropical & extreme tropical conditions.
The chamber shall be of mono block structure construction, comprising three cabinet placed one above the other (Hot cabinet at top, ambient cabinet at middle and cold cabinet at bottom).
Exterior : The exterior housing must be made from lightweight, self-supporting, corrosion resistant, galvanized steel-sheet of suitable thickness. The outside surface of the exterior wall shall be primed and painted to a standard finish with high quality materials.
Interior : Complete construction of the inner test spaces shall be made of stainless steel AISI 304 or equivalent materials. All the joints shall be continuous seam welded to form a hermetically sealed unit in order to prevent external moisture migration.
Upper test cabinet shall be hot cabinet and equipped with fan, heaters, temperature sensor, thermostat etc. Middle test cabinet shall be ambient cabinet and equipped with fan, heaters, temperature sensor, thermostat etc. The lower test cabinet shall be cold cabinet and shall equipped with fan, heaters, temperature sensor, thermostat etc. All the fan motors shall be placed outside the test space with suitable protection. If cooling by LN 2 injection method an appropriately sized port shall be provided at the center of the cold cabinet wall to vent the vaporized LN 2 so as to avoid over pressurization in the test space. Materials used in test space conditioning shall not generate any corrosive elements like sulphur, chlorine etc. Temperature sensor or any other items in the test cabinet shall not protrude into the usable test space volume of specimen carriage basket.
The chamber shall have door at front side. The door shall be fully open able (180 Degree C or more) to facilitate test article loading and unloading. Heaters shall be provided at the door at appropriate place to avoid moisture condensation on the outside surface. Suitable numbers of locks / clamps shall be provided to ensure the uniform pressure and air tightness of the chamber test space.
The chamber door shall have multi-pane observation window of size > 0.3m x 0.3m to view/inspect the test specimen in the ambient cabinet. the test space shall have proper lighting for functional inspection.
A backpressure free automatic drain shall be provided at cold cabinet and ambient cabinet to remove humidity / condensation water.
Licensed operating system with CD media. Licensed GUI based software for control, monitoring with visual flow diagram for chamber parameters (Temperatures of all three cabinet) visualization (Installation software to be provided on CD media). Computer communication, networking, data-logging software. All parametric data of temperature should be logged preferably in a database for archival and analysis.
Program management for production cycles and test. Easy creation of test programs. Looping features to allow repeating complete or partial programs. Guaranteed soak feature to let process variable reach set point before going to the next step. Operation of the chamber shall be by entering set points for immediate execution or by entering sequence of steps (set points, times, rates etc.) for timed automatic operation.
All the wires shall be suitably numbered or labeled for easy identification and documentation. The wires shall be strapped / bundled and routed through the ducts. All electrical circuits shall have reliable safety device like fuse or circuit breaker. Power connection terminal block and ground lugs shall be provided for easy utility connection. Complete wiring shall meet International Standard Code.
Electric connection according to IEC specification. Smooth starting of electrical system. Electrical protection against load faults such as short circuit, overload etc. Electrical protection against supply faults such as single phasing, low voltage and negative phase sequence. Safety push button in case of emergency.
ISO 9001:2015 Certified Co. D&B registered & CE mark product.
Calibration & validation certificates traceable to NABL / ERTL accredited labs.
Please note that this equipment has to be operated at an ambient temperature of +25ºC +/- 2ºC & suitable rating Servo Controlled Voltage Stabilizer has to be installed in case of voltage fluctuations.
To comply with the documentation requirements, we provide IQ, OQ & PQ protocols to be executed before taking the chamber in to for regular use, and support the supply with the following documents with the detail operational and service manual. Standard Operating Procedure (SOP) Operational Manual for Controller IQ,PQ,OQ protocols certificate. Calibration certificate of all controlling modules with traceability. Certificate of MOC. Test Report of chamber prior to supply with mapping certificate. wiring diagram for ease of service maintenance.